With these circuits you can make an insulation tester going upto two tera ohm, hence currents will be in pico amps, great care required in design. The rotary switches for voltages and ganged interlocked range switches must not be phenolic but industrial epoxy based molded switches.
The parts list which is not in the circuit, is listed below, the circuit is 15 years old. but you may get some idea on high resistance measurement. Use 1% MFR for all Resistors and low leakage plastic caps for low values. These instruments i used to calibrate with Victoreen Resitors and a Electrometer from Princeton Research, I faintly remember.
Read more »